- Open Access
Harmonic scalpel versus flexible CO2 laser for tongue resection: A histopathological analysis of thermal damage in human cadavers
© Hanby et al; licensee BioMed Central Ltd. 2011
- Received: 1 May 2011
- Accepted: 1 August 2011
- Published: 1 August 2011
Monopolar cautery is the most commonly used surgical cutting and hemostatic tool for head and neck surgery. There are newer technologies that are being utilized with the goal of precise cutting, decreasing blood loss, reducing thermal damage, and allowing faster wound healing. Our study compares thermal damage caused by Harmonic scalpel and CO2 laser to cadaveric tongue.
Two fresh human cadaver heads were enrolled for the study. Oral tongue was exposed and incisions were made in the tongue akin to a tongue tumor resection using the harmonic scalpel and flexible C02 laser fiber at various settings recommended for surgery. The margins of resection were sampled, labeled, and sent for pathological analysis to assess depth of thermal damage calculated in millimeters. The pathologist was blinded to the surgical tool used. Control tongue tissue was also sent for comparison as a baseline for comparison.
Three tongue samples were studied to assess depth of thermal damage by harmonic scalpel. The mean depth of thermal damage was 0.69 (range, 0.51 - 0.82). Five tongue samples were studied to assess depth of thermal damage by CO2 laser. The mean depth of thermal damage was 0.3 (range, 0.22 to 0.43). As expected, control samples showed 0 mm of thermal damage. There was a statistically significant difference between the depth of thermal injury to tongue resection margins by harmonic scalpel as compared to CO2 laser, (p = 0.003).
In a cadaveric model, flexible CO2 laser fiber causes less depth of thermal damage when compared with harmonic scalpel at settings utilized in our study. However, the relevance of this information in terms of wound healing, hemostasis, safety, cost-effectiveness, and surgical outcomes needs to be further studied in clinical settings.
- Laser Fiber
- Thermal Damage
- Harmonic Scalpel
- Oral Tongue
- Hypopharyngeal Squamous Cell Carcinoma
Data values and Descriptive statistics
Thermal depth in mm
Data Entry and Statistical Analysis
A Microsoft Excel Spreadsheet and Statistical Package for the Social Science version 13.0 was maintained for the data entry and statistical analysis. Thermal depth between harmonic scalpel and CO2 laser was compared using Independent sample T-test. A p-value less than 0.05 was considered statistically significant.
The CO2 laser was first introduced into Otolaryngology by Jako and Strong in 1972 . At a wavelength of 10,600 nm, this laser is rapidly absorbed by the water in the tissues ensuring minimal thermal damage and spread. This makes the CO2 laser particularly well suited for use near critical anatomical structures . Historically, the rapid absorption of this long wavelength laser by all known materials prevented its transmission via a flexible fiber. Therefore, most surgical CO2 lasers are applied via a bulky articulating arm either attached to a handpiece or to a micromanipulator mounted on an operating microscope. In this manner, the CO2 laser has been well established in the management of early glottic, supraglottic, oral and oropharyngeal and hypopharyngeal squamous cell carcinoma . In areas where exposure is limited such as the posterior oropharynx, the bulky delivery system and the inability to use visualize the area being resected in the surgeon's line of sight have been factors that have limited a more wide spread use of this technology for this indication. Similarly, the laser can be an excellent tool for anterior oral cavity and anterior oral tongue resections. However, the bulk and cumbersome delivery system makes its use less attractive [3, 7]. With the advent of the photonic band gap fiber assembly (PBFA), a flexible fiber CO2 delivery system developed by Omniguide Inc, it is now possible to overcome these limitations [6, 7]. The PBFA system allows the direct delivery of CO2 energy to regions in the head and neck where direct visualization is limited. A variety of hand pieces allow laser energy to be provided along the plane of surgical dissection and in sync with the surgeon's line of sight. This facilitates precise surgery. In addition to increased maneuverability, a variable rate of gas is transmitted through the hollow core of the PBFA creating the added benefits of cooling the surgical site and clearing the field of debris, plume and blood [2, 7].
The current limitations of the flexible CO2 laser fiber include a learning curve associated with its use in terms of maximizing its effectiveness. The PBFA also can be damaged if not used correctly. Although the tip of the fiber can provide tactile feedback to the surgeon, it is not robust enough to serve as a surgical dissector [1, 2]. Another known limitation of the standard CO2 laser was that it was inefficient with respect to coagulation (vessels up to 1-2 mm in diameter). The new PBFA fiber can easily be focused to improve cutting and also defocused to coagulate by moving the tip of the laser fiber closer to the target or away from it1. However, a true assessment of the lasers utility and ease of use for coagulation can only be derived from clinical studies.
The harmonic scalpel is able to cut and coagulate at a lower temperature (max 150°C) using mechanical vibration at 55,500 cycles per second [3, 8]. The harmonic scalpel like the laser causes less degree of thermal damage as compared to the monopolar cautery and has the ability to coagulate larger diameter vessels as compared to the laser ( 5 mm vs. 1-2 mm) which can be important in surgical resection of the tongue; an area that has a rich vascular supply [3, 8].
As surgeons, we are all on a perpetual search for a perfect cutting tool. The ideal instrument would accomplish the necessary functions of cutting and coagulation while minimizing collateral tissue injury. Decreasing overall tissue injury has obvious clinical implications with regard to surgical precision and less obvious clinical implications with regard to outcomes measures like expediency in return to work and normal diet. The CO2 laser and the harmonic scalpel have proven to be superior to monopolar electrocautery in minimizing collateral tissue damage.
In a cadaveric model, our study showed that flexible PFBA CO2 laser fiber causes less depth of thermal damage when compared with harmonic scalpel at recommended settings at the surgical margin. While it is tempting to extrapolate these findings into potential clinical benefits, further clinical studies are necessary to compare both surgical tools in terms of wound healing, hemostasis, safety, cost-effectiveness, and surgical outcomes.
The authors declare that they have no competing interests.
The authors would like to thank Anthony Wells and Reginald Delmore from the LSU Department of Anatomy for their help with the human cadaver specimens, Adam Hurst (OmniGuide systems, Inc) for his help with providing the Flexible CO2 laser fiber system, and Ethicon Endosurgery for providing us with the Harmonic scalpel and system for our experiment.
Omni-Guide Laser Systems
- Wolf T, Bhattacharyya S: Synopsis of results: comparison of tongue incisions created using the flexible CO2 laser fiber and monopolar electrosurgery in an animal model. Head and Neck White paper archive. 2010Google Scholar
- Pons Y, Gauthier J, Clement P, Conessa C: Ultrasonic partial glossectomy. Head Neck Oncol. 2009, 24;1: 21-View ArticleGoogle Scholar
- Devaiah AK, Shapshay SM, Desai U, Shapira G, Weisberg O, Torres DS, Wang Z: Surgical utility of a new carbon dioxide laser fiber: functional and histologic study. Laryngoscope. 2005, 115 (8): 1463-8. 10.1097/01.mlg.0000171021.73635.3b.View ArticlePubMedGoogle Scholar
- Solares CA, Strome M: Transoral robot-assisted CO2 laser supraglottic laryngectomy: experimental and clinical data. 2007, 117 (5): 817-20.Google Scholar
- Zeitels SM, Kobler JB, Heaton JT, Faquin W: Carbon dioxide laser fiber for laryngeal cancer surgery. Ann Otol Rhinol Laryngol. 2006, 115 (7): 535-41.View ArticlePubMedGoogle Scholar
- OmniGuide Inc. http://www.omni-guide.com
- Holsinger FC, Prichard CN, Shapira G, Weisberg O, Torres DS, Anastassiou C, Harel E, Fink Y, Weber RS: Use of the photonic band gap fiber assembly CO2 laser system in head and neck surgical oncology. Laryngoscope. 2006, 116 (7): 1288-90. 10.1097/01.mlg.0000227557.61978.18.View ArticlePubMedGoogle Scholar
- Energy Devices Overview. http://www.harmonic.com
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.